Applying the OMPS (Ozone Mapping and Profiler Suite) derived calibration to EPIC's counts per second to obtain Earth albedos for each of the UV channels, the reflectances can be used to retrieve total column ozone (TCO), Lambert equivalent reflectivity (LER), aerosol optical depth and absorption, aerosol index (AI), and UV reflectance at the Earth's surface. Ozone retrieval requires the use of measured laboratory high spectral resolution absorption coefficients. The EPIC measured reflectance spectra are compared with a set of radiative transfer derived lookup tables for the EPIC filter transmission functions and for a wide range of ozone values. LER, AI, and ozone are retrieved simultaneously with a maximum resolution of 18 km at the sub-satellite point. A matched pair of ozone and LER images are shown in Fig. A1 for August 21, 2016 at 16:58 UT.
EPIC ozone has been compared to ozone retrieved from a Pandora Spectrometer Instrument (PSI) located in Boulder Colorado matched in location and time several times per day. The average agreement is 2.1 ± 5.4% (Fig. A2). An additional comparison (Fig. A3) has been made with the assimilated ozone product from the Modern-Era Retrospective Analysis for Research and Applications, version 2 (MERRA-2) based on Microwave Limb Sounder (MLS) and total column ozone from the Ozone Monitoring Instrument (OMI). All of the structures in the EPIC ozone retrieval are present in the MERRA-2 assimilation model ozone, but with an average offset of about 3%. Comparisons with MERRA-2 have been made using ozone data from other satellites that have similar offsets.
The EPIC Ozone products are available at the Atmospheric Science Data Center (ASDC) at NASA Langley Research Center: https://search.earthdata.nasa.gov/search/collections?q=dscovr_epic_l2_to3_01&ok=dscovr_epic_l2_to3_01
Figure A1. EPIC retrieved ozone and LER values for April 21, 2016 at 16:58 UTC. The ozone scale is from 100 to 500 DU, and the LER scale is from 0 to 100 percent.
Figure A2. Left: EPIC ozone data compare to Pandora retrievals at Boulder Colorado. Right: daily (grey circles) and monthly (solid line) average difference between Pandora and EPIC ozone retrievals.
Figure A3. Comparison of EPIC total column ozone with the MERRA-2 assimilation ozone model for April 17, 2016.